We have a new computer running Windows 10 64bits.
In order to use Winspec, we installed it on a Virtual Box running Windows XP 32bits. The software is then working and we can communicate with the monochromator and the camera.
But we have a huge problem: we cannot decrease the exposure time below 0.01s because then we get directly the error message "Data overrun due to hardware conflict".
As proposed in the manual we changed the DMA buffer value but it doesn't change anything.
The problem seems to be linked to the use of the virtual box.
Do you have any idea that could help ?
Thank you in advance,
EDIT : Changing from Fast Mode to Safe Mode in Acquisition -> Experiment Setup -> Timing solve the problem !!